-Good Image Quality — Part 1 -Reliability – Part two -Price – Element 3 -Functionality : Part 4
Portion 4 Functionality
-Eva by simply Dent-X (AFP Imaging) -SuniRay-Suni Corporation (Orange Dental) -Dixi 3-Planmeca -Dexis Platinum-Dexis -CDR Elite-Schick -Kodak RVG6100- Kodak (Owned By simply Carestream Now)
Component 4: Functionality
In this article I wish to be able to discuss what sort of fÃ¼hler can be connected into your personal computer systems, what application the sensors will certainly work with in addition to any particular files that might get of interest in terms of the sensors make use of.
-Eva by Dent-X (AFP Imaging)
This sensor connects for an inexpensive and durable software that can always be mounted on the particular wall or under a counter. The interface connects to the computer through USB. The interface inside this design will be not intended in order to be moved coming from room to space and because on this the interface is sold rather inexpensively at around $300. The particular interface itself could be 16 feet in the computer because of the limitations associated with USB. I possess not really had good luck using attaching this user interface through a powered extension. To end up being sure of good stability I would guide to limit you to ultimately mounting the user interface within the 16ft limitation. The fÃ¼hler then plugs straight into the interface plus give you around 6 feet associated with cable to operate with. If the space has move across xray tube heads We would advice going the usb cable through the wall structure or floor into your move across therefore that when an individual plug-in the sensor, the sensor in addition to sensor cable follow the same course as the xray tube arm. This will prevent you by tripping on the wire and damaging the sensor during work with. If 16 feet will not get you to the pass through you may want to mount the particular unit at your 13 o’clock. Understand what need a pass via than try in order to route your hardware cable over to the particular xray tube wall membrane mount and track down your interface generally there. The idea is the fact if you usually are using the fÃ¼hler, the tube mind is pulled away plus the arm regarding the tube mind keeps you from walking across where your own sensor cable is definitely.
The sensor is definitely of similar dimensions to a Suni Ray or Kleidsam sensor and has got rounded corners. We would expect this specific sensor to end up being as comfortable seeing that most sensors about the market.
Avoi sensors come using their own involved imaging software which is okay to work with or if a person would like these people also work perfect in Xray Eye-sight. I prefer Xray Vision in most cases, because of its potential to work using many different sensors, panos and phosphor platter systems.
-SuniRay-Suni Company (Orange Dental)
The particular Suni Ray is a direct inside of line USB sensor similar to the particular Kodak sensor. Typically the sensor is attached to its very own in-line interface plus has about 3 feet of cable here. The little user interface about the size of a tin regarding mints has a mini USB women connection and comes with an of sixteen foot usb wire to plug in to the interface. This messfÃ¼hler design is meant to be plugged into a powered center located near your computer. In most of my installs I suggest having good enough USB cables in order to have one connected every room and just move the messfÃ¼hler with its quick cable and software around. This is usually much easier when compared to the way using the very long 16 foot usb cable and moving it from room to room.
The particular sensor has curved corners plus a composite body. sensor cable types have found the particular sensor to be secure to use and even position.
Suni Beam sensors work inside of Prof Suni and even Xray Vision. This specific is type of redundant because Prof Suni is an ORIGINAL EQUIPMENT MANUFACTURING release of Xray Vision
The Planmeca sensor has an interface that the actual messfÃ¼hler plugs into. The particular interface has many different ways being connected to typically the computer including UNIVERSAL SERIAL BUS and a system interface. The community interface is the excellent choice due to the fact you can spot your interface any distance from the computer you decide on. When you used the correct wireless system in conjunction with the interface and even you know a new computer guy who is able to set it right up you wouldn’t even need to any cable from your network to typically the interface.
The fÃ¼hler itself is water-proof and of related size to typically the other sensors inside this review apart from the Dexis which can be smaller.
Planmeca detectors work in their particular own Dimaxis computer software and can also be utilized in Xray Eyesight
The new Dexis Platinum does not have any interface. It has a wire that comes directly out of typically the sensor and finishes in a regular USB B Connection that can be plugged into the computer UNIVERSAL SERIES BUS ports. The disadvantage of this is that typically the connector is just excellent for about 1000 disconnects meaning after about 1000 patients you will most likely start having issues along with the sensor plug if you are really unplugging it and moving it right after each patient. The cable is fairly long and many likely would have to get plugged in anywhere your computer is definitely and draped across the floor to be able to your patient when in use.
Dexis sensors only work within Dexis Image from the things i have recently been told although I know that motorists were made with regard to it to be effective throughout Xrayvision for when the sensor is used in conjunction with the US army.
Kleidsam Sensor Plug Intended for Interface
This messfÃ¼hler plugs in through its own interface which includes another cable television that comes out of it in addition to plugs in via USB to the particular computer. As significantly as placement associated with the sensor an individual will probably desire to place the user interface on the computer place as you wouldn’t need to damage the particular cable visiting the interface by running this through your walls. The interfaces regarding these schick receptors are near $2K a piece so be careful together with them.
Schick devices work in Dentrix Image through a new module you download, through eaglesoft photo and with xray eye-sight. I have only employed the sensor within eaglesoft and xray vision. If you use eaglesoft image you might like to head out into the adjustments and change the purchase timer. By standard I believe it is set to significantly less than a minute so if you hit the acquire button before positioning the sensor typically the acquisition may moment out before a person take those xray.
-Kodak RVG6100- Kodak (Owned By Carestream Now)
Kodak RVG 6100 MessfÃ¼hler
Kodak devices are very similar in order to the Suni Beam as to just how they plug-in. These people have an interface that is quite small which is attached directly to typically the sensor through the cable. Then a person use an HARDWARE cable to plug into the software. As far as functionality the Kodak works similar to the Suni Ray.
Kodak receptors work in Kodak Image and in Xray Vision. Kodak only recently introduced the drivers to interface with Xray Vision when you have Kodak Image and don’t like it feel free to buy Xray Vision and move over.